[ Identification | Description | Input parameters | Links ]

# The `ILL_IN6` Instrument

The IN6 Time-of-Flight simulation at the ILL (instrument only).

## Identification

- Site: ILL
- Author: Emmanuel Farhi
- Origin: ILL (France)
- Date: 17th Jan 2005.

## Description

    IN6 is a time focussing time-of-flight spectrometer designed for quasielastic and
    inelastic scattering for incident wavelengths in the range of 4 to 6 Angs.

    An intense beam is extracted from the H15 guide by a vertically focussing
    monochromator array. It consists of three composite pyrolytic graphite
    monochromators using the full height (20 cm) of the guide and focussing the beam
    at the sample position. In order to minimise the interference with the
    subsequent instruments, the monochromator can deliver only four wavelengths:
    4.1; 4.6; 5.1; and 5.9 Angs. The second order reflection from the graphite
    monochromator is removed by a beryllium-filter cooled at liquid nitrogen
    temperature.
    To achieve the time-focussing condition, the beam is pulsed by a Fermi chopper.
    It has a small slot length to ensure a good transmission. The normal distance
    between the Fermi chopper and the sample is 38 cm. To prevent frame-overlap when
    the chopper is rotating faster than 7500 rpm, a suppressor chopper is placed
    before the Fermi chopper and rotates in phase with the latter.

    The secondary spectrometer consists first of an evacuated sample area. The
    detector bank is entirely covered with detector boxes, thus avoiding the
    inconvenience of moving the counters.

    This instrument model contains a cold source a triple monochromator
    (using the GROUP), two Fermi Choppers (including one background chopper), a
    liquid sample handling coherent and incoherent processes (elastic and inelastic)
    with multiple scattering, customized monitors, and the SPLIT mechanism to
    improve the statistics. The H15 guide is not described in this model.

## Input parameters

Parameters in boldface are required; the others are optional.

Name | Unit | Description | Default
lambda | Angs | wavelength within 4.14|4.6|5.12|5.92 | 4.14
dlambda | Angs | wavelength HALF spread. default is 0.075 | 0.075
SPEED | rpm | Fermi chopper speed. -1=auto, 0=stopped in open pos. | -1
M1 | coder values | monochromator motor 1 position. -1=auto | -1
M2 | coder values | monochromator motor 2 positinn. -1=auto | -1
M3 | coder values | monochromator motor 3 position. -1=auto | -1
MONITOR | something like time in s | monitor preset | 1
CHA_WIDTH | us | channel width. -1=auto | -1
TOF_DELAY | us | TOF delay. -1=auto | -1
TOF_CHA_RESOL | 1 | number of channels. | 128
ELPEAK | 1 | elastic peak channel. -1=auto | -1
RATIO | 1 | Suppressor speed ratio. -1=no suppressor. | 1
mFC | 1 | super mirror FermiChopper coating m-value | 0
PHASE | deg | Fermi phase w/r/ to Suppressor. -360=auto | -360
Sqw_coh | str | sample coherent S(q,w) file name | "Rb_liq_coh.sqw"
Sqw_inc | str | sample incoherent S(q,w) file name | "Rb_liq_inc.sqw"
radius | m | outer radius of sample hollow cylinder | 0.01
thickness | m | thickness of sample hollow cylinder | 0.005

## Links

- Source code for `ILL_IN6.instr`.
- The IN6@ILL Yellow Pages
- R.Scherm et al, "Another Time of Flight Spectrometer", ILL Report 76S235, 1976
- R.Scherm, "A high-resolution spectrometer ...", report Jul-295-NP, Kernforschungsanlage Julich, 1965
- Y.Blanc, "Le spectrometre a temps de vol IN6", ILL Report 83BL21G, 1983
- K.H.Beckurts et al, Neutron physics, Kernforschungsanlage Karlsruhe, 1964 (p317)
- R.Scherm and T.Springer, "Proposal of a multiple Chopper...", Kernforschungsanlage Julich, 19xx

[ Identification | Description | Input parameters | Links ]